COD Medium Range Reagent Vials, EPA Method (25 tests) - HI93754B-25
6 850,49 Lekë
The HI93754B-25 are COD reagent vials that follow the EPA approved method for the medium-range determination of chemical oxygen demand (COD) using a compatible benchtop photometer. These high quality reagents are manufactured in our state-of-the-art facility and are clearly marked with the lot number and expiration date on each box for traceability
- Pre-dosed reagents for ease of use
- Supplied with certificate of quality
- Marked with expiration date and lot number for traceability
The HI93754B-25 COD kit contains 25 ready-to-use vials containing pre-measured reagents. These high-quality reagents follow the EPA approved method 410.4. In this method, heated digestion is required to oxidize organic matter in the sample. During the digestion, hexavalent chromium in the reagent is reduced to trivalent chromium. The amount of reduced chromium is then measured colorimetrically. The intensity of color is determined by a compatible photometer and the COD concentration is displayed in mg/L (ppm) O2. These reagents are designed to be used with samples that have an expected range of 0 to 150 mg/L (ppm) O2.
Hanna vials contain approximately 3 mL of pre-dosed reagent so the user simply needs to add a small quantity of the sample. With pre-dosed vials, test preparation time is dramatically reduced and there is no time-consuming reagent preparation procedure or glassware cleaning. The vials and caps of the COD reagents have been designed to avoid accidental reagent spills. Due to the pre-dosed reagents, the amount of chemicals and handling time is also minimized.
Comparison of MethodsEPA- The EPA approved method 410.4 for COD determination requires the use of mercuric sulfate (HgSO4). Mercuric sulfate is present in the EPA reagent vials to mitigate the interference of chloride in the measurement sample. This is a common requirement for wastewater treatment plants and food manufacturing facilities which tend to have higher concentrations of chloride present in their effluent. When using reagents that follow the EPA method, it is important to treat used vials as hazardous waste, as mercury waste requires recycling. Dichromate - The dichromate method is adapted from the standard EPA and ISO methods for determination of COD, which are approved to measure COD concentrations ranging up to 1500 mg/L O2. For samples with expected concentrations in the high range of 0 to 15000 mg/L O2, the dichromate reagents can be used for accurate COD determination. It is important to treat used vials as hazardous waste, as mercury waste requires recycling. Mercury-free - For those who do not wish to handle mercury reagents, our “green,” mercury-free reagents are available for COD determination. These reagents are more sensitive to chloride interferences, as mercuric sulfate is not present. While the mercury-free reagents are easier to dispose of, they may be unacceptable for reporting purposes. ISO- The ISO approved method 15705 for COD determination also requires the use of mercuric sulfate (HgSO4) to reduce the interference of chloride. Containing similar chemical reagents as the EPA method, it is important to treat used ISO vials as hazardous waste, as mercury waste requires recycling.
|Method||Parameter||Range (as O2)||Resolution||Accuracy||Reagent COD|
|EPA||COD LR COD MR||0 to 150 mg/L 0 to 1500 mg/L||1 mg/L 1 mg/L||±5 mg/L ±5% of reading ±15 mg/L ±4% of reading||HI93754A-25 HI93754B-25|
|Dichromate||COD HR||0 to 15000 mg/L||10 mg/L||±150 mg/L ±3% of reading||HI93754C-25|
|Mercury-free||COD LR COD MR||0 to 150 mg/L 0 to 1500 mg/L||1 mg/L 1 mg/L||±5 mg/L ±5% of reading ±15 mg/L ±4% of reading||HI93754D-25 HI93754E-25|
|ISO||COD LR COD M||0 to 150 mg/L 0 to 1500 mg/L||1 mg/L 1 mg/L||±5 mg/L ±5% of reading ±15 mg/L ±4% of reading||HI93754F-25 HI93754G-25|