Pool Line Cleaning Solution 25 sachets - HI7006014P
- 25 sachets each containing 20 mL of solution
- Convenient packaging for one time use
- Ideal for infrequent use
The most common cause for pH measurement inaccuracies is an unclean or improperly cleaned electrode. This is very important to note, because during calibration, the instrument assumes that the electrode is clean and that the standardization curve created during the calibration process will remain a valid reference until the next calibration. The pH meters on the market today will allow an offset voltage (pH 7.01) of approximately ±60 mV. The deviation from 0 mV is not unusual but ideally should be no greater than ±30 mV. The calibration process compensates for the change in offset voltage. If the mV offset continues to deviate and the meter is calibrated then any changes to the probe will result in inaccurate readings. Cleaning is a fast and effective routine that should be performed on a regular basis as a preventative measure against using a dirty electrode and to ensure that the junction is not clogged. By simply immersing the electrode in the sachet containing the cleaning solution for 15 to 20 minutes will dissolve any mineral deposits or other coatings. After any cleaning procedure is performed it is important to rehydrate the glass of a pH electrode by placing in storage solution for 15 minutes. After the probe is rehydrated then a calibration must be performed. This is very important for process probes since many are used to monitor and control the dosing of chemicals. It is critical that the probe is clean and calibrated in order for a meter to maintain the ideal set point. The cleaning of pH and ORP process probes should be part of a routine maintenance program.
Light block foil packaging
- Light-tight packet helps to prevent against degradation by UV light
One-time use sachet packets
- Ensures quality and freshness of solution
Each sachet marked with lot number and expiration date
- Hanna cleaning solutions are specially formulated to have an expiration of 5 years from manufacture for an unopened sachet.
|Description||General purpose cleaning solution|